The Hardware card shows circle diagrams or level bars for every input channel, reflecting the live amplitudes of incoming signals. Both examples are shown in the following figures. For convenience, they can be grouped per sensor type and/or Virtual Point. Unmapped channels can be shown or hidden.
Signal amplitudes shown as circle diagrams
Signal amplitudes shown as level bars
Channels are auto-zeroed by right-clicking in the Hardware card and selecting Auto-zero all Channels. It is recommended to auto-zero all channels any time new equipment is connected, when coming back from coffee breaks, or any other unattended interval.
Once the DAQ is connected, the measurement is automatically set to active such that the DAQ is ready for a trigger. The measurement can be paused by pressing the Pause Measurement within the Hardware card. The key command for this is F10. The status is shown in the taskbar.
Contact us for more VIBES
Contact our support team or call us on +31 85 744 09 70
Subscribe to our newsletter
Privacy statement. By submitting this form you agree to having your details registered in our database. At your request, we'll remove your details from our database immediately *